Wind Speed and Wind Direction sensors
The Wind Speed and Wind Direction sensors can be connected to the universal D105 meter.
Dashboard with examples widgets of monitoring wind speed and direction below »
There are two types of sensors.
1. With the digital output via RS485 interface
Wind speed and wind direction sensors transmit information via the RS485 bus. This interface is notable for its stable working at large distances, up to 100 meters. Through the RS485-1Wire interface, which comes in a box from smart-MAC only, the sensors are connected to the digital input (4) of the smart meter smart-MAC D105.
In the settings of the D105 device select the type of sensor as "Temperature", in the "Counter" section.
After saving the settings, the device will search for connected sensors.
Match the connected wind sensors with the corresponding data parameter T1 - T5 and give them a name.
For the wind speed sensor, the data will be in meters per second.
For the wind direction sensor, the data will contain a number from 0 to 15, i.e. the sensor has 16 graduations depending on the direction angle in the measuring range 0 .. 360 °. Where 0 corresponds to 0 degrees of the North direction, and each step is to 360 ° / 16 = 22.5 °.
It is necessary to position the sensor North by the mark on the sensor.
In the widget settings, in the section "Arithmetic operation", select multiply by 22.5 and get the angle for the direction of the wind.
2. With Analog output signal 0 .. 5V / 0 .. 10V / etc.
Such a sensor is connected to the analog input (5) ADC of the smart-MAC D105 device.
Wind Speed sensor - Anemometer with Analog Output
To calculate wind speed, you need to know the range of speed measurement, for example, 0 - 30 m/s and the range of voltage variation at the analog input, for example, 0 - 5V.
Given the linear relationship between wind speed and voltage at the analog output of the sensor, it is easy to calculate the coefficient for converting voltage to speed. To do this, divide the speed measurement range by the voltage measurement range. For our example:
K = 30 / 5 = 6
In the widget settings select the [ADC] Analog data and in the "Arithmetic operation" section multiply by 6.
Done, now the widget will display the wind speed in meters per second.
Wind Direction sensor with Analog Output
To calculate the direction of the wind, you need to know the range of voltage measurement at the analog input, for example, 0-5V, in accordance with the direction. An output voltage of 0V corresponds to 0 degrees North. And it is necessary to position the sensor North by the mark on the sensor. The wind direction sensor has 16 voltage gradations depending on the direction angle in the measurement range 0 .. 360 °.
We can calculate it:
Voltage step = 5V / 16 = 0.3125V Direction angle step = 360° / 16 = 22.5° Wind direction value = ADC / 0.3125 * 22.5 = ADC * (22.5 / 0.3125 ) = ADC * 72
The coefficient for converting the voltage into the direction angle is 72.
In the widget settings select the [ADC] Analog data and in the "Arithmetic operation" section multiply by 72.
Done, now the widget will display the wind direction in the degree 0..360 °.
Everything done, you can return to Dashboard and monitor the wind!
Customer support service by UserEcho